UVA UVB Solarmeter Distributor
SOLAR METER 5
UV Lamp Aging Tests
Acrylic Shield Transmission
Eye wear UV Block Tests
Compare Browning Rays
Estimate Tt Time10
|Solar meter Specifications |
|Irrad. Range||0-199.9 mW/cm2 Total UV|
|Response||280-400 nm (UVB thru UVA)|
|Conv. Rate||3.0 Readings/Sec|
|Display||3.5 Digit LCD|
|Digit Size||0.4 inch high|
|Oper. Temp||32 F to 100 F|
|Oper. Humid.||5% to 80% RH|
|Dimensions (in.)||4.2L x 2.4W x 0.9D (in.)|
|Weight||4.5 OZ. (incl. batt.)|
|Power Source||9-Volt DC Battery|
|Lens||UVT Acrylic .030|
|Diffuser||Virgin PTFE .010|
CONTACT US by email for details.
Model 5.0 HP > 0-1999 mW/cm
(For high pressure lamps over 199.9 mW/cm)
Model 5.0 SF > 0-199.9 W/m
(For science fair projects)
Hamamatsu #G5842 UV SENSOR/FILTER
The G5842 is a semiconductor UV sensor consisting of a GaAsP photodiode chip and a UV filter. The G5842 is completely insensitive to visible light longer than 400 nm and infrared radiation since its spectral response covers only the UV region from 260 to 400 nm. Applications of the G5842 include solar UV detection, as the spectral response is well matched to the solar UV spectrum, and tanning lamps peaking near 365 nm ("new era" fluorescent and "high pressure" HID).
Press and hold push-button switch on face of unit. Aim sensor window in top panel of meter directly at UV source. Note reading on LCD and record if desired.
Battery operation voltage is 9V down to 6.5V. Below 6.5V the LCD numbers will begin to dim, indicating the need for battery replacement. Under "typical" service load, the battery should last about 2 years.
No. 001, Priya Enclave, Beside E-Seva, Road No. 9, Snehapuri Colony, Nacharam, Hyderabad, Telangana, 500076, India